The PT-500 is a rugged submersible pressure transmitter designed for highly accurate water level measurement in streams, lakes, rivers, and other water bodies.
The APG PT-500 submersible pressure transducer is designed for use in a wide variety of environmental monitoring applications. The sensor is ideally suited for measuring water level in weirs, flumes, small to large streams, ponds, lakes, irrigation channels, and groundwater wells. The sensor is installed in a fixed location below the minimum expected water level, and a cable containing the sensor signals and vent tube is connected to a data logger mounted above flood stage.
The PT-500 measures the combined pressure exerted on it by the atmosphere and the head of water above it. A vent tube in the cable automatically corrects for changes in barometric pressure, and measured values are mathematically compensated for all linearity and temperature errors. With +/-0.1% Full Scale accuracy, the PT-500 provides high performance and accuracy over a wide range of operating conditions. The housing is constructed with laser welded 316L stainless steel.
The sensor features a plug-and-play interface to NexSens iSIC data loggers via RS-485 Modbus digital output. Data collection options include direct-connect, landline phone, cellular, radio, Ethernet, WI-FI, and satellite telemetry. Each iSIC data logger simultaneously supports 8 analog inputs & 4 digital inputs for multi-sensor data logging capability. NexSens iChart Software is a Windows-based program for interfacing both locally (direct-connect) and remotely (through telemetry) to an iSIC data logger or network of data loggers.
|Image||Part #||Product Description||Price||Stock||Order|
|549014-1155||PT-500 submersible pressure transducer with RS-485 Modbus output, 6 PSIG range, 0.1% FS accuracy, 35 ft. cable||
|549014-1244||PT-500 submersible pressure transducer with RS-485 Modbus output, 6 PSIG range, 0.1% FS accuracy, 65 ft. cable||
|549014-1756||PT-500 submersible pressure transducer with RS-485 Modbus output, 6 PSIG range, 0.1% FS accuracy, 100 ft. cable||
Largely seen as pristine and relatively untouched by human activity thanks to its protected status, the portion of the Colorado River flowing through Grand Canyon National Park is anything but, according to recently published research. This is evidenced by high levels of selenium and mercury found in the fishes there. Scientists from many institutions were involved in the years-long work, full results of which have been published in the journal Environmental Toxicology and Chemistry. It was led by the U.S. Geological Survey, but perhaps the contributors from Idaho State University got the best end of the stick. They were looking into the food webs of the river to evaluate concentrations of selenium and mercury gathering in fish.Read More
For all the straightforward groundwater monitoring applications that the folks at Heron Instruments help with, there are a few that are far from typical. These include projects that take place near remediation sites or not far from waste disposal operations. Realizing that customers working in those sorts of projects are in need of a more robust option, the company has released the dipper-Tough . The new water level meter takes inspiration from Heron’s popular dipper-T , while throwing in a host of improvements that environmental pros working in groundwater can really appreciate.Read More
Scientists at Ohio State University are at the fore of the fight against harmful algal blooms in Lake Erie. In fact, they deployed a new cellular data buoy off the shore of Gibraltar Island in 2014, months before the Toledo Water Crisis spurred a boom in monitoring platforms around the lake. That was in part because researchers at the university’s Stone Laboratory, backed by Ohio Sea Grant and housed on Gibraltar, had been seeing a resurgence of blooms in the lake long before international attention came around following the crisis. There was an opportunity, they saw, to continue advancing the mission of research, education and outreach on Lake Erie. The cellular data buoy complimented that in a great way.Read More