The ATI Model Q46/84 Hydrogen Peroxide Monitor is designed to continuously monitor processes that utilize hydrogen peroxide for bleaching or disinfection purposes.
Hydrogen peroxide (H2O2) is an extremely strong oxidizer widely used in bleaching applications in the paper industry. It has also been used in a variety of applications including disinfection, odor control, oxygenation, and cyanide oxidation. It is frequently used in wastewater collection systems to remove hydrogen sulfide that destroys concrete pipes and manhole structures. Peroxide applications in aqueous systems, like most chemical treatment processes, function most efficiently with accurate measurement and control.
ATI’s Model Q46/84 Hydrogen Peroxide Monitor is designed to continuously measure the concentration of H2O2 in aqueous systems. Using a direct peroxide sensor, the unit responds rapidly to changes in concentration, enabling operators to control chemical feed to maintain specific targets. With both digital and analog communications available, the Q46 is adaptable to a wide variety of peroxide monitoring applications.
Hydrogen Peroxide sensors are amperometric devices isolated from the water by a peroxide permeable membrane. As peroxide diffuses through this membrane, it comes in contact with an active electrode and is oxidized on the surface. The net effect is that the sensor generates a current proportional to H2O2 concentration. An integral RTD in the sensor provides for automatic temperature compensation, allowing the sensor to operate accurately over a range of 0-50°C.
Peroxide sensors are capable of operating over a fairly broad concentration range, from a low range of 0-2 PPM up to a high range of 0-200 PPM. Systems will operate with good sensitivity down to 0.05 PPM and respond to changes in concentration within about 60 seconds, making them useful for automatic control.
Hydrogen Peroxide sensors require a steady flow of sample across the membrane at the tip of the sensing assembly. ATI offers options for flowcells, including the standard constant-head overflow system, and a sealed flowcell for pressures up to 50 PSI. For simplicity of installation, complete flow control assemblies are available. Panel assemblies are available with or without a flow switch and can greatly reduce installation time.
|Image||Part #||Product Description||Price||Stock||Order|
|Q46/84||Hydrogen peroxide monitor||Drop ships from manufacturer|
A complex series of locks and dams up and down the Ohio River enable interstate commerce, travel and recreation by maintaining a usable pathway for watercraft, but come with the inevitable byproducts of disrupting the river’s natural systems. To combat this, the U.S. Army Corps of Engineers uses a complex monitoring and response technology designed to minimize the negative impacts of dredging on the river ecosystem. Steven Foster, a limnologist with the Corps Water Quality Team, works at the Robert C. Byrd Lock and Dam in Gallipolis Ferry, West Virginia. He said one key area he focuses on is the welfare of mussels in the river. River dredging can smother mussel beds, so Foster and the team of engineers monitor the beds to ensure their safety.Read More
While the scientific community has formed its consensus on how ice sheets are shrinking in and around Greenland, some researchers are tracking what happens to the meltwater as it drains into the ocean each summer. Their study, published in Nature Geoscience by an interdisciplinary team of biologists, oceanographers and hydrologists, used computer models to simulate the meltwater to see where currents take it and what effect it could have on the ocean. Renato Castelao, one of the researchers and an associate professor of marine science for the University of Georgia, said one of the biggest discoveries of the study was the surprising final destinations of the ice sheets as they melt into the ocean each summer.Read More
Thinking of hitting the ice with a SondeCAM underwater fishing camera? Due to its rugged design, you won't have to worry about it handling the harsh elements. However there are a few simple tricks to get the most out of a FishSens SondeCAM while ice fishing. You won't have to do anything to modify the SondeCAM itself, but you are going to have to bring a few extra things. Most importantly we are going to need a power source. Unless you are hauling your gear with a truck, you'll want something more portable than the battery you used in the boat. Pick up an inexpensive and maintenance-free 12-volt, 9-amp battery. It is going to provide plenty of power, but will be much lighter and take up less space.Read More