ATI Model Q46/85 is designed for on-line monitoring and control of processes that contain peracetic acid.
Peracetic acid (PAA) is an extremely strong oxidizer widely used in the food industry for disinfection of piping systems and processing equipment. It is also used for spray washing of food products, and for disinfection of cooling water systems. As a disinfecting agent, PAA is often preferred because it produces no harmful breakdown products.
As with any disinfection system, maintaining proper residual values is the key to effective pathogen control. To facilitate reliable chemical feed control, ATI has developed an on-line monitor cable of providing real time measurement of low levels of PAA in solution. The Q46/85 Peracetic Acid Monitor uses a direct sensing polarographic probe. The probe is typically used in a flowcell, but a submersible sensor is also available. A permeable diffusion membrane isolates the sensing electrodes from the measured sample, providing long-term stability without electrode fouling problems. The measurement is selective for PAA and is not affected by changes in hydrogen peroxide concentration, which is always present in PAA solutions.
Peracetic acid sensors are amperometric devices isolated from the water by a permeable membrane. As PAA diffuses through this membrane, it comes in contact with an active electrode and is reduced on the surface. The net effect is that the sensor generates a current proportional to PAA concentration. An integral RTD in the sensor provides for automatic temperature compensation, allowing the sensor to operate accurately over a range of 0-50° C.
Peracetic acid sensors are capable of operating over a fairly broad concentration range, from a low range of 0-20 PPM up to a high range of 0-2000 PPM. Systems will operate with good sensitivity down to 0.2 PPM and respond to changes in concentration within about 60 seconds, making them useful for automatic control.
Peracetic Acid sensors require a steady flow of sample across the membrane at the tip of the sensing assembly. ATI offers options for flowcells, including the standard constant-head overflow system, and a sealed flowcell for pressures up to 50 PSI. For simplicity of installation, complete flow control assemblies are available. Panel assemblies are available with or without a flow switch and can greatly reduce installation time. For open tank installation, a submersible sensor is available. Sample agitation is required as PAA sensors cannot be used in still water.
|Image||Part #||Product Description||Price||Stock||Order|
|Q46/85||Peracetic acid monitor||Drop ships from manufacturer|
An understanding of climate change’s effects on the environment has become commonplace and grows every day, but one researcher from Florida State University is looking to answer a new question: What are climate change’s effects on people’s health? In one of the first studies of its kind, Chris Uejio, an assistant professor at FSU, and a team of researchers studied how climate change can affect the roughly 20 million Americans (according to the Environmental Protection Agency) who consume untreated drinking water on a daily basis. Because climate forecasts are predicting higher rainfall rates over the next few decades, coming down in intense storms, Uejio said those flashes could cause flare-ups in waterborne illnesses.Read More
We put together this infographic on data buoys for our Spring 2017 edition of the Environmental Monitor ( PDF available online ). Organizations across the globe use data buoy systems to observe and monitor atmospheric and oceanographic conditions in remote locations. Measurements range from air pressure, humidity, wind speed and direction to wave height, water temperature, dissolved oxygen and other water quality parameters. With the help of national and international networks, reliable and comprehensive data sets are made available for research and public safety.Read More
The Spring 2017 Environmental Monitor is on the way to subscribers this month. Our quarterly print editions feature the best of the Monitor's coverage from the past few months with added photos, graphics, updates and the latest monitoring gear. If you don't have a print subscription, you can sign up for free. If you'd like to peruse some of our past editions, check out our print archive . In this edition, we showcase a number of projects that are truly advancing the way data are gathered in the environmental monitoring field. This includes a look at the first-ever deployment of the ESPniagara in Lake Erie, a device for real-time microcystin measurements that is so advanced its makers say it is essentially a robot.Read More