ATI Model Q46/85 is designed for on-line monitoring and control of processes that contain peracetic acid.
Peracetic acid (PAA) is an extremely strong oxidizer widely used in the food industry for disinfection of piping systems and processing equipment. It is also used for spray washing of food products, and for disinfection of cooling water systems. As a disinfecting agent, PAA is often preferred because it produces no harmful breakdown products.
As with any disinfection system, maintaining proper residual values is the key to effective pathogen control. To facilitate reliable chemical feed control, ATI has developed an on-line monitor cable of providing real time measurement of low levels of PAA in solution. The Q46/85 Peracetic Acid Monitor uses a direct sensing polarographic probe. The probe is typically used in a flowcell, but a submersible sensor is also available. A permeable diffusion membrane isolates the sensing electrodes from the measured sample, providing long-term stability without electrode fouling problems. The measurement is selective for PAA and is not affected by changes in hydrogen peroxide concentration, which is always present in PAA solutions.
Peracetic acid sensors are amperometric devices isolated from the water by a permeable membrane. As PAA diffuses through this membrane, it comes in contact with an active electrode and is reduced on the surface. The net effect is that the sensor generates a current proportional to PAA concentration. An integral RTD in the sensor provides for automatic temperature compensation, allowing the sensor to operate accurately over a range of 0-50° C.
Peracetic acid sensors are capable of operating over a fairly broad concentration range, from a low range of 0-20 PPM up to a high range of 0-2000 PPM. Systems will operate with good sensitivity down to 0.2 PPM and respond to changes in concentration within about 60 seconds, making them useful for automatic control.
Peracetic Acid sensors require a steady flow of sample across the membrane at the tip of the sensing assembly. ATI offers options for flowcells, including the standard constant-head overflow system, and a sealed flowcell for pressures up to 50 PSI. For simplicity of installation, complete flow control assemblies are available. Panel assemblies are available with or without a flow switch and can greatly reduce installation time. For open tank installation, a submersible sensor is available. Sample agitation is required as PAA sensors cannot be used in still water.
|Image||Part #||Product Description||Price||Stock||Order|
|Q46/85||Peracetic acid monitor||Drop ships from manufacturer|
Largely seen as pristine and relatively untouched by human activity thanks to its protected status, the portion of the Colorado River flowing through Grand Canyon National Park is anything but, according to recently published research. This is evidenced by high levels of selenium and mercury found in the fishes there. Scientists from many institutions were involved in the years-long work, full results of which have been published in the journal Environmental Toxicology and Chemistry. It was led by the U.S. Geological Survey, but perhaps the contributors from Idaho State University got the best end of the stick. They were looking into the food webs of the river to evaluate concentrations of selenium and mercury gathering in fish.Read More
For all the straightforward groundwater monitoring applications that the folks at Heron Instruments help with, there are a few that are far from typical. These include projects that take place near remediation sites or not far from waste disposal operations. Realizing that customers working in those sorts of projects are in need of a more robust option, the company has released the dipper-Tough . The new water level meter takes inspiration from Heron’s popular dipper-T , while throwing in a host of improvements that environmental pros working in groundwater can really appreciate.Read More
Scientists at Ohio State University are at the fore of the fight against harmful algal blooms in Lake Erie. In fact, they deployed a new cellular data buoy off the shore of Gibraltar Island in 2014, months before the Toledo Water Crisis spurred a boom in monitoring platforms around the lake. That was in part because researchers at the university’s Stone Laboratory, backed by Ohio Sea Grant and housed on Gibraltar, had been seeing a resurgence of blooms in the lake long before international attention came around following the crisis. There was an opportunity, they saw, to continue advancing the mission of research, education and outreach on Lake Erie. The cellular data buoy complimented that in a great way.Read More