ATI’s Model Q46/88 Suspended Solids Monitor provides real time monitoring of suspended solids in a variety of water and wastewater applications.
ATI’s Model Q46/88 Suspended Solids Monitor provides real time monitoring of suspended solids in a variety of water and wastewater applications. A submersible sensor immersed in process tanks or effluent channel senses particulates in the water using an optical backscatter technique that allows measurement over a wide range. Results are displayed on the Q46 electronic unit mounted near the sensor with a variety of outputs provided as standard.
Monitoring suspended solids in wastewater and industrial process water can be useful for either process control or for alarming of unusual conditions. In biological treatment systems, monitoring suspended solids in the aeration tank can assist operators in maintaining optimum MLSS (Mixed Liquor Suspended Solids) concentration. In industrial clarifier's, suspended solids water quality monitoring can warn of upset conditions that might result in the discharge of solids that exceed plant permits.
Suspended solids sensors are optical devices operating in the infrared region. Unlike turbidity sensors that use 90 degree scatter to optimize sensitivity, suspended solids sensors use “backscatter” to allow solids measurements at much higher levels. Operation with infrared light ensures very long sensor life and minimizes the effects of changing sample color.
Sensors are designed to withstand the rigorous conditions of wastewater and industrial process streams and to last for years of service with nothing more than occasional cleaning of the sensing surface. There are no protruding surfaces near the sensing element to avoid accumulation of fibrous materials. The sensor is simply pipe mounted using mounting adapters available from ATI.
Optical sensors used for monitoring biologically active systems such as aeration tanks or aerobic digestors will require periodic cleaning to maintain the integrity of the measurement. Biological slime deposited on the optical surface will degrade the ability to transmit IR light into the sample. The frequency of cleaning varies widely depending on the turbulence in the process. Course bubble diffusion systems tend to scour the sensor while fine bubble diffusion systems result in more rapid sensor fouling.
Cleaning can be done manually by simply wiping the sensor as needed, but ATI also offers an automatic air-blast cleaning system as an option. The “Q-Blast” air cleaning system is controlled by the Q46/88 Suspended Solids Monitor and provides a compact air compressor system that periodically applies pulses of compressed air across the optical surface to remove accumulated biofouling. This system greatly reduces the requirement for manual maintenance, with cleaning frequency programmed to occur as often as necessary.
|Image||Part #||Product Description||Price||Stock||Order|
|Q46/88||Suspended solids monitor||Usually ships in 1-2 weeks|
Thinking of hitting the ice with a SondeCAM underwater fishing camera? Due to its rugged design, you won't have to worry about it handling the harsh elements. However there are a few simple tricks to get the most out of a FishSens SondeCAM while ice fishing. You won't have to do anything to modify the SondeCAM itself, but you are going to have to bring a few extra things. Most importantly we are going to need a power source. Unless you are hauling your gear with a truck, you'll want something more portable than the battery you used in the boat. Pick up an inexpensive and maintenance-free 12-volt, 9-amp battery. It is going to provide plenty of power, but will be much lighter and take up less space.Read More
We've all felt the frustration of weeding through a school of dinks to catch a "keeper." Often the small fish outnumber the bigger ones and they are typically more aggressive. Sometimes there's no choice but to deal with it, as is often the case with open water fishing. However a frozen lake involves a vertical presentation and a stable platform, it's a perfect situation to pick and choose which fish you want. Once you locate a school and get set up it's time to start sizing them up with a FishSens SondeCAM underwater fishing camera. It can be mind-blowing just how big some of these schools of fish are and also how outnumbered fish of a desirable size can be.Read More
It’s no secret that anglers have been the means by which invasive species and non-native fish have spread to new water bodies in the past. Fishermen have even been known to transport some of their favorite fish to new areas on purpose so that they can catch them a little closer to home. And the results of those actions have not always been ideal. In Ontario, Canada, fishermen have taken non-native bass and stocked them into what were historically lakes dominated by brook and cutthroat trout. The actions have impacted ecosystems, but scientists have been unable to broadly study the effects because they didn’t have enough data. But that is no longer the case for some Ontario lakes, as a study from biologists at the University of Toronto shows.Read More