The Q46T Monitor is designed to meet the needs of both municipal and industrial water treatment for reliable, low-range turbidity measurement.
ATI’s Model Q46T Turbidity Monitor is designed to meet the needs of both municipal drinking water systems and industrial water treatment for reliable, low-range turbidity measurement. Using an infrared or energy source in this 90 degree scatter measurement, the system provides high sensitivity measurement with unmatched zero stability. Turbidity measurements down to 0.001 NTU or as high as 2000 NTU can be measured with the same monitor, eliminating the need for separate high and low range instruments.
The turbidity sensor used in the Q46T monitor is a planer sensor with the light source and photo detector mounted on a flat face. Lenses in front of the light source direct the beam of light at a 45-degree angle into the sample. Another lens in front of the photo detector collects the 90-degree scattered light and directs it to the detector. The signal generated by the detector is amplified inside the sensor for transmission to the display unit. Periodic pulsing of the light source allows the sensor zero to be adjusted automatically for improved stability in low-range applications. Turbidity sensors are available for measurement in a pressurized flowcell or for direct submersion in open tanks. A 1-1/2” flow T assembly is also available for direct in-line measurements. The flowcell is required for very low measurements (0-2 NTU), while the submersion unit may be used for higher turbidity applications such as wastewater effluent or raw water.
Air bubbles are a common problem in many turbidity systems. ATI’s turbidity sensor is designed to operate under pressure to eliminate the sample degassing that often causes air bubble errors. Sample pressure remains constant until the sample exits the measuring chamber, resulting in a more reliable measurement. The turbidity electronics unit is an AC or DC powered instrument providing an LCD display of turbidity value, as well as indication of alarm status and instrument diagnostics. The NEMA 4X can be wall, pipe, or handrail mounted using the universal mounting kit included with each system.
The monitor provides display of turbidity over a variety of operating ranges. The minimum display range of 0-2 NTU provides resolution down to 0.001 NTU and is suitable for almost any final filter monitoring application. Ranges of 0- 20 or 0-200 NTU are available for raw water or clarifier effluent monitoring. The Q46T provides three programmable alarms with SPDT alarm relays are included, as well as two isolated 4-20 mA outputs. The analog outputs may be programmed for full-scale outputs as low as 0-0.2 NTU, and can be inverted if desired.
Q46T monitors also provide sensor diagnostic functions to warn of condition that cause inaccurate or invalid readings. The sensor is continuously monitored for optical fouling, and will display an alarm message when the sensor requires cleaning In addition, the sensor will detect the lack of water in the flowcell and provide a “dry cell” when an air interface is detected. These alarm conditions will cause a third alarm relay to activate, which can be used to indicate these conditions remotely.
ATI’s Q-Blast option provides the ideal answer for submersible turbidity applications such as wastewater effluent. Employing a unique “air-blast” cleaning method, sensors can be cleaned as often as necessary without operator attention. Pulses of pressurized air delivered through a nozzle at the tip of the sensor remove accumulated solids from critical sensing surfaces, resulting in accurate and reliable measurements.
The Q-Blast Auto-Clean assembly is housed in a NEMA 4X enclosure suitable for indoor or outdoor use. The system includes an integral compressor and air-pulse control components, with a power supply for the entire air supply system incorporated into the design. A simple connection to the Q46T monitor provides the sequencing for the system and allows the operator to select cleaning frequencies as often as once every hour to as little as once every 999 hours. To insure performance in extreme cold conditions, a thermostatically controlled heater is included in the assembly, allowing operation down to -40°C.
|Image||Part #||Product Description||Price||Stock||Order|
|Q46T||Turbidity monitor||Drop ships from manufacturer|
A complex series of locks and dams up and down the Ohio River enable interstate commerce, travel and recreation by maintaining a usable pathway for watercraft, but come with the inevitable byproducts of disrupting the river’s natural systems. To combat this, the U.S. Army Corps of Engineers uses a complex monitoring and response technology designed to minimize the negative impacts of dredging on the river ecosystem. Steven Foster, a limnologist with the Corps Water Quality Team, works at the Robert C. Byrd Lock and Dam in Gallipolis Ferry, West Virginia. He said one key area he focuses on is the welfare of mussels in the river. River dredging can smother mussel beds, so Foster and the team of engineers monitor the beds to ensure their safety.Read More
While the scientific community has formed its consensus on how ice sheets are shrinking in and around Greenland, some researchers are tracking what happens to the meltwater as it drains into the ocean each summer. Their study, published in Nature Geoscience by an interdisciplinary team of biologists, oceanographers and hydrologists, used computer models to simulate the meltwater to see where currents take it and what effect it could have on the ocean. Renato Castelao, one of the researchers and an associate professor of marine science for the University of Georgia, said one of the biggest discoveries of the study was the surprising final destinations of the ice sheets as they melt into the ocean each summer.Read More
Thinking of hitting the ice with a SondeCAM underwater fishing camera? Due to its rugged design, you won't have to worry about it handling the harsh elements. However there are a few simple tricks to get the most out of a FishSens SondeCAM while ice fishing. You won't have to do anything to modify the SondeCAM itself, but you are going to have to bring a few extra things. Most importantly we are going to need a power source. Unless you are hauling your gear with a truck, you'll want something more portable than the battery you used in the boat. Pick up an inexpensive and maintenance-free 12-volt, 9-amp battery. It is going to provide plenty of power, but will be much lighter and take up less space.Read More