The Levelogger artesian well fitting makes monitoring of artesian conditions straightforward.
Monitoring of artesian conditions in which the piezometric surface is above ground surface or more particularly above the top of well casing elevation using Leveloggers can be quite straight forward.
Continuous artesian conditions infer that the piezometric surface never drops below the level the ground surface or particularly the top of casing elevation and the casing is sealed with a sealed wellhead. In this case, where freezing is not a concern, the Levelogger need only be installed on the wellhead itself by means of a large compression fitting with a 1/4" NPT connector. Solinst can supply a 7/8" nylon compression fitting for this purpose. First, a 1/4" NPT hole is tapped into the wellhead. The user slides the Levelogger into the compression fitting, leaving about 1/4 of the logger body exposed above the fitting. The compression fitting nut is tightened around the Levelogger. The 1/4" NPT connector on the bottom of the pressure chamber of the fitting threads into the hole in the sealed wellhead. The user can communicate with the logger simply by removing the logger cap and attaching the optical reader. Ensure that the logger and sealed wellhead are enclosed within an outer protective well cap or enclosure. This method should only be used in low pressure conditions (less than 30 psi or 66 ft (20 m) of water column).
When conducting artesian monitoring with Leveloggers a number of considerations must be kept in mind. First, ensure that the maximum hydraulic pressure the Levelogger will encounter within the well at its installation point will not exceed the hydraulic range of the logger. Second, artesian conditions do not preclude the necessity for barometric compensation of Levelogger data. Artesian conditions are caused by aquacludes forming confined aquifers. Confined aquifers, while not acted on by barometric pressure to the same extent as unconfined aquifers, are typically subject to barometric pressure at some barometric efficiency. Finally, bear in mind that the total pressure and subsequent water column equivalent depth measured by the Levelogger after barometric compensation may not represent the actual water level within the artesian well. Sealed intermittent artesian wells can be pressurized when artesian, but can also be de-pressurized when non artesian. The Levelogger’s reading after barometric compensation represent the height of the piezometric surface.
|Image||Part #||Product Description||Price||Stock||Order|
|112350||Sealed artesian well connector assembly||
|Usually ships in 1-2 weeks|
For decades, commercial fishing for yellow perch was allowed in southern Lake Michigan. This persisted until 1996 when it was outlawed, giving perch stocks there some time to recover. Scientists had for some time assumed that this fishing ban would not affect the reproduction cycles of the perch quickly and that they were going to need a long time to revert back to the cycles they relied on before commercial fishing ever started. But new research led by scientists at Purdue University finds that maturation schedules of yellow perch in southern Lake Michigan are much more resilient than had been previously thought possible.Read More
Largely seen as pristine and relatively untouched by human activity thanks to its protected status, the portion of the Colorado River flowing through Grand Canyon National Park is anything but, according to recently published research. This is evidenced by high levels of selenium and mercury found in the fishes there. Scientists from many institutions were involved in the years-long work, full results of which have been published in the journal Environmental Toxicology and Chemistry. It was led by the U.S. Geological Survey, but perhaps the contributors from Idaho State University got the best end of the stick. They were looking into the food webs of the river to evaluate concentrations of selenium and mercury gathering in fish.Read More
For all the straightforward groundwater monitoring applications that the folks at Heron Instruments help with, there are a few that are far from typical. These include projects that take place near remediation sites or not far from waste disposal operations. Realizing that customers working in those sorts of projects are in need of a more robust option, the company has released the dipper-Tough . The new water level meter takes inspiration from Heron’s popular dipper-T , while throwing in a host of improvements that environmental pros working in groundwater can really appreciate.Read More