ATI Q46/76 Turbidity Monitor
- Optional Q-Blast removes build-up on the sensor, reducing sensor maintenance
- Contact outputs include two programmable control relays for control and alarm modes
- Communication Options for Profibus-DP, Modbus-RTU, or Ethernet-IP
|Q46/76||Turbidity monitor|| |
|Drop ships from manufacturer|
ATI’s Model Q46/76 Turbidity Monitor is designed to meet the needs of both municipal drinking water systems and industrial water treatment for reliable, low-range turbidity measurement. Using an infrared or energy source in this 90 degree scatter measurement, the system provides high sensitivity measurement with unmatched zero stability. Turbidity measurements down to 0.001 NTU or as high as 2000 NTU can be measured with the same monitor, eliminating the need for separate high and low range instruments.
The turbidity sensor used in the Q46/76 monitor is a planer sensor with the light source and photo detector mounted on a flat face. Lenses in front of the light source direct the beam of light at a 45-degree angle into the sample. Another lens in front of the photo detector collects the 90-degree scattered light and directs it to the detector. The signal generated by the detector is amplified inside the sensor for transmission to the display unit. Periodic pulsing of the light source allows the sensor zero to be adjusted automatically for improved stability in low-range applications. Turbidity sensors are available for measurement in a pressurized flowcell or for direct submersion in open tanks. A 1-1/2” flow T assembly is also available for direct in-line measurements. The flowcell is required for very low measurements (0-2 NTU), while the submersion unit may be used for higher turbidity applications such as wastewater effluent or raw water.
Air bubbles are a common problem in many turbidity systems. ATI’s turbidity sensor is designed to operate under pressure to eliminate the sample degassing that often causes air bubble errors. Sample pressure remains constant until the sample exits the measuring chamber, resulting in a more reliable measurement. The turbidity electronics unit is an AC or DC powered instrument providing an LCD display of turbidity value, as well as indication of alarm status and instrument diagnostics. The NEMA 4X can be wall, pipe, or handrail mounted using the universal mounting kit included with each system.
The monitor provides display of turbidity over a variety of operating ranges. The minimum display range of 0-2 NTU provides resolution down to 0.001 NTU and is suitable for almost any final filter monitoring application. Ranges of 0- 20 or 0-200 NTU are available for raw water or clarifier effluent monitoring. The Q46/76 provides three programmable alarms with SPDT alarm relays are included, as well as two isolated 4-20 mA outputs. The analog outputs may be programmed for full-scale outputs as low as 0-0.2 NTU, and can be inverted if desired.
Q46/76 monitors also provide sensor diagnostic functions to warn of condition that cause inaccurate or invalid readings. The sensor is continuously monitored for optical fouling, and will display an alarm message when the sensor requires cleaning In addition, the sensor will detect the lack of water in the flowcell and provide a “dry cell” when an air interface is detected. These alarm conditions will cause a third alarm relay to activate, which can be used to indicate these conditions remotely.
ATI’s Q-Blast option provides the ideal answer for submersible turbidity applications such as wastewater effluent. Employing a unique “air-blast” cleaning method, sensors can be cleaned as often as necessary without operator attention. Pulses of pressurized air delivered through a nozzle at the tip of the sensor remove accumulated solids from critical sensing surfaces, resulting in accurate and reliable measurements.
The Q-Blast Auto-Clean assembly is housed in a NEMA 4X enclosure suitable for indoor or outdoor use. The system includes an integral compressor and air-pulse control components, with a power supply for the entire air supply system incorporated into the design. A simple connection to the Q46/76 monitor provides the sequencing for the system and allows the operator to select cleaning frequencies as often as once every hour to as little as once every 999 hours. To insure performance in extreme cold conditions, a thermostatically controlled heater is included in the assembly, allowing operation down to -40°C.
In The News
As we hear more and more about algal blooms of different kinds across the United States, teams of scientists are working hard to ensure that they don't become our new normal. One project in Florida is taking a multi-disciplinary approach to the problem—including genetic analysis.
The team's work is part of a full-court press in Florida recently, making a serious push to understand what is triggering more frequent blooms. Jose Lopez, Ph.D. , of Nova Southeastern University , the primary investigator on the genetic analysis portion of the project, spoke to EM about the project and his work on it.
“This is a very good project,” explains Dr. Lopez. “We're excited about it, and it's a lesson in persistence.”
From extreme weather such as Hurricane Harvey to spills and other accidents, the Gulf Coast of Texas is no stranger to dangerous situations. This is where the data provided by the Texas Automated Buoy System ( TABS ) comes into the picture.
Among the nation's most successful and longest-running coastal ocean-observing systems at the state level, the TABS real-time oceanographic buoy system monitors currents, waves, salinity, winds, and other parameters. Dr. Anthony Knap , director of Geochemical Environmental Research Group (GERG) and a Professor of Oceanography at Texas A&;M University, spoke to EM about working with TABS.
“TABS has been running now for 24 years,” explains Dr. Knap.Read More
Formed by a glacier, Jordan Pond is among Maine's clearest, most beautiful bodies of water. It's also a critical freshwater resource, and watchful eyes are protecting it.
EM spoke with Dr. Rachel Fowler, Friends of Acadia's aquatic scientist, about her work monitoring Jordan Pond. A postdoctoral research scientist at the University of Maine, she is a member of a partnership among the National Park Service, the University of Maine Climate Change Institute, and Friends of Acadia that began deploying the Jordan Pond buoy in 2013. Canon provided the initial support for the project.
Friends of Acadia is a nonprofit organization that supports different projects in the park.Read More