The Q46/76 Monitor is designed to meet the needs of both municipal and industrial water treatment for reliable, low-range turbidity measurement.
ATI’s Model Q46/76 Turbidity Monitor is designed to meet the needs of both municipal drinking water systems and industrial water treatment for reliable, low-range turbidity measurement. Using an infrared or energy source in this 90 degree scatter measurement, the system provides high sensitivity measurement with unmatched zero stability. Turbidity measurements down to 0.001 NTU or as high as 2000 NTU can be measured with the same monitor, eliminating the need for separate high and low range instruments.
The turbidity sensor used in the Q46/76 monitor is a planer sensor with the light source and photo detector mounted on a flat face. Lenses in front of the light source direct the beam of light at a 45-degree angle into the sample. Another lens in front of the photo detector collects the 90-degree scattered light and directs it to the detector. The signal generated by the detector is amplified inside the sensor for transmission to the display unit. Periodic pulsing of the light source allows the sensor zero to be adjusted automatically for improved stability in low-range applications. Turbidity sensors are available for measurement in a pressurized flowcell or for direct submersion in open tanks. A 1-1/2” flow T assembly is also available for direct in-line measurements. The flowcell is required for very low measurements (0-2 NTU), while the submersion unit may be used for higher turbidity applications such as wastewater effluent or raw water.
Air bubbles are a common problem in many turbidity systems. ATI’s turbidity sensor is designed to operate under pressure to eliminate the sample degassing that often causes air bubble errors. Sample pressure remains constant until the sample exits the measuring chamber, resulting in a more reliable measurement. The turbidity electronics unit is an AC or DC powered instrument providing an LCD display of turbidity value, as well as indication of alarm status and instrument diagnostics. The NEMA 4X can be wall, pipe, or handrail mounted using the universal mounting kit included with each system.
The monitor provides display of turbidity over a variety of operating ranges. The minimum display range of 0-2 NTU provides resolution down to 0.001 NTU and is suitable for almost any final filter monitoring application. Ranges of 0- 20 or 0-200 NTU are available for raw water or clarifier effluent monitoring. The Q46/76 provides three programmable alarms with SPDT alarm relays are included, as well as two isolated 4-20 mA outputs. The analog outputs may be programmed for full-scale outputs as low as 0-0.2 NTU, and can be inverted if desired.
Q46/76 monitors also provide sensor diagnostic functions to warn of condition that cause inaccurate or invalid readings. The sensor is continuously monitored for optical fouling, and will display an alarm message when the sensor requires cleaning In addition, the sensor will detect the lack of water in the flowcell and provide a “dry cell” when an air interface is detected. These alarm conditions will cause a third alarm relay to activate, which can be used to indicate these conditions remotely.
ATI’s Q-Blast option provides the ideal answer for submersible turbidity applications such as wastewater effluent. Employing a unique “air-blast” cleaning method, sensors can be cleaned as often as necessary without operator attention. Pulses of pressurized air delivered through a nozzle at the tip of the sensor remove accumulated solids from critical sensing surfaces, resulting in accurate and reliable measurements.
The Q-Blast Auto-Clean assembly is housed in a NEMA 4X enclosure suitable for indoor or outdoor use. The system includes an integral compressor and air-pulse control components, with a power supply for the entire air supply system incorporated into the design. A simple connection to the Q46/76 monitor provides the sequencing for the system and allows the operator to select cleaning frequencies as often as once every hour to as little as once every 999 hours. To insure performance in extreme cold conditions, a thermostatically controlled heater is included in the assembly, allowing operation down to -40°C.
|Image||Part #||Product Description||Price||Stock||Order|
|Q46/76||Turbidity monitor||Drop ships from manufacturer|
Growing from a 38-acre purchase in 1998 to 298 acres in 2004 to the 305 acres it encompasses today; the Black Fork River Wetlands features habitats not found just anywhere, including buttonbush swamp, swamp forest, marsh, riparian corridor and uplands habitats. Beavers make their homes there, as well as trumpeter swans, bald eagles, soras and sandhill cranes.
While it may seem picturesque and undisturbed, it is in fact embattled due to human activity on all sides. “It’s a multi-use area,” says Jenna Binder, a visiting Assistant Professor in Ashland University’s Biology and Toxicology Department. “It’s strongly influenced by the heavy agriculture in this area of Ohio. Oil and gas industry fracking is also being done in the area.Read More
Biological field stations make it possible for researchers all over the country to conduct environmental research. While some field stations have artist residencies, art is typically not the main focus of the biological station. Not so at Bakersville, North Carolina’s new AS IF Center (Art + Science In The Field) , which just opened its doors in March 2018. At AS IF, researchers and artists are deliberately invited to commingle, collaborate and create new things together. Far from being on the periphery or existing as an afterthought, artists are considered to be on parity with researchers at AS IF, the one energized by the other’s perspective.Read More
The polar regions of the world have always a challenge for scientists to explore and study. Even logistics that are typically no more than passing concerns under other circumstances such as transportation become major problems during polar wintertime. Now, r esearchers are reporting on their use of hundreds of oceanic floats that are drifting and diving their way through the Southern Ocean, including under its ice, with surprising results.
Happy robotic wanderers
EM spoke with Dr. Alison Gray , assistant professor of physical oceanography at the University of Washington , to find out more about the work, the robots, and the significance of the findings in improving our understanding of the global climate and this poorly studied region.Read More