ATI Q46/85 Peracetic Acid Monitor
- Available with constant head or sealed flowcell configuration
- Contact outputs include two programmable control relays for control and alarm modes
- Communication Options for Profibus-DP, Modbus-RTU, or Ethernet-IP
|Q46/85||Peracetic acid monitor|| |
|Drop ships from manufacturer|
Peracetic acid (PAA) is an extremely strong oxidizer widely used in the food industry for disinfection of piping systems and processing equipment. It is also used for spray washing of food products, and for disinfection of cooling water systems. As a disinfecting agent, PAA is often preferred because it produces no harmful breakdown products.
As with any disinfection system, maintaining proper residual values is the key to effective pathogen control. To facilitate reliable chemical feed control, ATI has developed an on-line monitor cable of providing real time measurement of low levels of PAA in solution. The Q46/85 Peracetic Acid Monitor uses a direct sensing polarographic probe. The probe is typically used in a flowcell, but a submersible sensor is also available. A permeable diffusion membrane isolates the sensing electrodes from the measured sample, providing long-term stability without electrode fouling problems. The measurement is selective for PAA and is not affected by changes in hydrogen peroxide concentration, which is always present in PAA solutions.
Peracetic acid sensors are amperometric devices isolated from the water by a permeable membrane. As PAA diffuses through this membrane, it comes in contact with an active electrode and is reduced on the surface. The net effect is that the sensor generates a current proportional to PAA concentration. An integral RTD in the sensor provides for automatic temperature compensation, allowing the sensor to operate accurately over a range of 0-50° C.
Peracetic acid sensors are capable of operating over a fairly broad concentration range, from a low range of 0-20 PPM up to a high range of 0-2000 PPM. Systems will operate with good sensitivity down to 0.2 PPM and respond to changes in concentration within about 60 seconds, making them useful for automatic control.
Peracetic Acid sensors require a steady flow of sample across the membrane at the tip of the sensing assembly. ATI offers options for flowcells, including the standard constant-head overflow system, and a sealed flowcell for pressures up to 50 PSI. For simplicity of installation, complete flow control assemblies are available. Panel assemblies are available with or without a flow switch and can greatly reduce installation time. For open tank installation, a submersible sensor is available. Sample agitation is required as PAA sensors cannot be used in still water.
In The News
For most humans, mayflies seem like a nuisance, hovering over the waterways as we try to enjoy them. However, for anyone hoping to monitor the health of watersheds, mayflies are important aquatic species—and now, a digital version of the mayfly is helping some scientists keep an eye on the water. Research scientist Dr. Scott Ensign , who serves as Assistant Director of the Stroud Water Research Center , spoke to EM about how the digital mayfly technology developed.
“ Shannon Hicks is the engineer who started developing the Mayfly six or seven years ago,” explains Dr. Ensign.Read More
Time is of the essence when it comes to tracking algal blooms, and people everywhere are looking for solutions. In Florida, scientists from Florida Atlantic University Harbor Branch Oceanographic Institute (HBOI) recently trialed a solar-powered, algae-tracking sail boat developed by Navocean , Inc. Dr. Jordon Beckler of Florida Atlantic University (FAU) directs HBOI's Geochemistry and Geochemical Sensing Lab and spoke to EM about the trials and the boat.
"This boat is so amazing when you see it in action," remarks Dr. Beckler. "Navocean originally contacted me a few years back about a demonstration when I was over at my previous institution in West Florida, and we brainstormed some scenarios for employing the boat for harmful algae bloom monitoring.Read More
CICHAZ Biological Field Station Provides A Unique Educational and Research Experience in Mexico’s Huasteca Region
The story of the Centro de Investigaciones Científicas de las Huastecas "Aguazarca" (CICHAZ) Biological Field Station, a member of the Organization of Biological Field Stations ( OBFS ), starts with Dr. Gil Rosenthal, Professor of Biology and Chair of Ecology and Evolutionary Biology at Texas A &; M University . Rosenthal has worked in the Huasteca region of Mexico since 1994 and for years kept his research equipment at a local ranch/hotel with the dream of one day having a field station where he could run experiments with collaborators and students. Since 2005, Rosenthal has been the Co-Director of the field station along with his wife, Dr.Read More