ATI Q46H/79S Total Chlorine Stripping Monitor
- Measurement is made without contact between water sample and sensor
- Contact outputs include two programmable control relays for control and alarm modes
- Communication Options for Profibus-DP, Modbus-RTU, or Ethernet-IP
|Q46H/79S||Total chlorine stripping monitor|| |
|Usually ships in 1-2 weeks|
Residual chlorine is found in many chemical forms in water systems. Residuals in clean water are often predominantly free chlorine while wastewater and cooling water can contain mixtures of free chlorine, combined chlorine, and organochlorine species. Measurement of residual chlorine in applications where only free chlorine (potable water) or only combined chlorine (chloraminated water) exist can often be monitored with direct sensor measurement. However, applications where a variety of chlorine forms can exist (wastewater effluent and some cooling water) require a more complicated measurement method. These applications generally require a “Total Chlorine” measurement and involve chemically converting all chlorine species into a single chemical form.
This is normally done by reacting the sample with pH 4 buffer and potassium iodide to convert various chlorine compounds into iodine. Many on-line monitors for total chlorine use this iodometric method, often measuring the current between two exposed electrodes to determine iodine concentration. ATI’s Model Q46H/79S uses this same standard iodometric chemistry, but with a unique sensing technique for measuring the released iodine. The system takes the reacted sample containing iodine and uses an air-stripping system to remove molecular iodine from solution. The gas-phase iodine from the water sample is channeled through a conditioning module and then directly to an iodine gas sensor. The result is that the iodine measurement is made without any contact between the water sample and the sensor. Contaminants in the sample that cause fouling and contamination of standard electrodes do not affect the Q46 system, providing greater operational reliability.
In The News
If someone speaks to Jesse Ellis, Assistant Professor of Biology at Coe College and Director of the Wilderness Field Station, they might get interrupted; by a blue-headed vireo.
“Bird songs are a big part of data gathering for research here,” says Ellis. “We use automated recording units (ARUs) to record wilderness sounds, especially sounds made by birds and frogs.”
The Wilderness Field Station is a teaching-oriented facility. “In addition to our annual summer classes, we also conduct bird studies here including bird counts in transects, and researchers from other colleges come here to do multiple lake samplings,” Ellis adds.Read More
For most humans, mayflies seem like a nuisance, hovering over the waterways as we try to enjoy them. However, for anyone hoping to monitor the health of watersheds, mayflies are important aquatic species—and now, a digital version of the mayfly is helping some scientists keep an eye on the water. Research scientist Dr. Scott Ensign , who serves as Assistant Director of the Stroud Water Research Center , spoke to EM about how the digital mayfly technology developed.
“ Shannon Hicks is the engineer who started developing the Mayfly six or seven years ago,” explains Dr. Ensign.Read More
Time is of the essence when it comes to tracking algal blooms, and people everywhere are looking for solutions. In Florida, scientists from Florida Atlantic University Harbor Branch Oceanographic Institute (HBOI) recently trialed a solar-powered, algae-tracking sail boat developed by Navocean , Inc. Dr. Jordon Beckler of Florida Atlantic University (FAU) directs HBOI's Geochemistry and Geochemical Sensing Lab and spoke to EM about the trials and the boat.
"This boat is so amazing when you see it in action," remarks Dr. Beckler. "Navocean originally contacted me a few years back about a demonstration when I was over at my previous institution in West Florida, and we brainstormed some scenarios for employing the boat for harmful algae bloom monitoring.Read More