Eno Scientific Well Sounder 2010 PRO Water Level Meter
- Automatic temperature compensation
- Calculate drawdown and recovery rates
- Data points stored in non-volatile memory
|2010P||Well Sounder 2010 PRO water level meter|
|5201-10||Probe extension cable, 10 ft|
|Usually ships in 3-5 days|
|5300||120 VAC plug-in adapter|
|Usually ships in 3-5 days|
A built-in data logger can record up to 13,000 time stamped readings along with the well ID, temperature, and other associated measurements. The logger can be set to take measurements automatically at intervals between 1 and 60 minutes, or manually by pressing the log button on the keypad. The recorded data can be reviewed on the screen or downloaded to a computer for analysis using the serial RS-232 port or the included USB cable. Stored data is formatted in simple Windows-compatible text files sorted by well ID. Data can be easily copied from the instrument onto the computer and opened with almost any common program such as Word, Excel, or Notepad.
Other features of the Well Sounder 2010 PRO includes a power save mode which puts the unit to sleep between measurements when logging and turns the power off after a selected period of inactivity. If there are known defects in the well such as a rock fissure, simply set the measurement range minimum and maximum to ignore the interference. The multi-purpose bi-directional serial port can also be used to transmit data in real-time to a remote readout or logging device.
- Internal Power: 6 AA Alkaline batteries
- External Power (Optional): 5.5 to 12VDC at 50ma
- Real-Time Clock: Li Ion 3V battery CR2032
- Resolution: .05 ft
- Accuracy: 0.1 ft
- Range: 9 to 2000 ft
- Temperature Compensation: Built-in to probe
- Memory: Non-volatile flash memory for 13,000 data points
- Logging Rates: 1 second to 60 minutes
- Temperature: -10 to 110 F
- Humidity: 10 to 90% non-condensing
- Dimensions: 4"x 7.5"x 1.25"
- Weight: ~13 oz
- Display: 2x16 character LCD
- USB port: For access to file system
- Serial Output: 300 to 57600 baud, 8 data bits, 1 stop bit (19200 baud default)
- (1) Well Sounder 2010 PRO meter
- (1) Probe with 6 ft cable
- (1) USB cable
- (1) Carrying case
- (1) US Groundwater Temperature Chart
- (1) Quick Start Guide
- (1) User Manual
The sonic water level meter unit and probe are resistant to rain and splashing but not saturation or submersion. They will be damaged if submerged.
The maximum range is 2000 feet. On a 6" well with tight clean casing all the way down and little to no obstructions along the way, the maximum depth can be achieved. On uncased wells drilled through porous stone or with irregular walls, or any obstructions such as spacer rings or couplings, the range will be reduced.
When the log memory is full, the sounder stops recording new data and preserves the first data points. The Well Sounder 2010P will continue to record data in its extended data memory up to 2GB and be accessible with the USB connection.
Hand dug wells are often fairly large and irregular, which may cause multiple or weak sound reflections. However, in any difficult to measure well, a small tube can be installed specifically for measurement. A 3/4" PE pipe can be used for up to 1000 feet provided splicers do not restrict the ID.
The Eno Scientific Well Sounder was designed to work on a closed well. A simple piece of rigid card board or plastic held against the opening is enough to meet this requirement. In many cases it will work on the open well but may be off by a couple feet.
Yes, the sonic water level meter can be used with six AA alkaline batteries or with a 6.5-12 VDC external power source. If using an external source, the power must not exceed 16 VDC.
In The News
Measuring the distance to the water’s surface in a well can be difficult, especially when the well is deep and full of obstacles that can block and entangle mechanical measuring tools.
But the Eno Scientific Well Sounder 2010 Pro gets around those obstacles with the power of sound, eliminating the need for any mechanical measuring. The device sends low-frequency sound waves down a well until they hit liquid.
“The sound can go around all the wiring and pipes and stabilizers within the well and give a measurement within a second,” said Rachel Bean, Eno Scientific sales manager.
The Well Sounder’s probe emits broadcasts low-frequency sound waves from the top of a well. The waves return in about one second per 500 feet.Read More
The Charles River used to be a swimming hotspot for Cambridge and Boston residents.
Decades of industrial pollution and nutrient runoff have degraded water quality and eliminated public swimming in the Lower Charles, but a movement is afoot to get Boston and Cambridge back in the water. One step toward the goal of a safely swimmable river—without the need to obtain a permit, as is now necessary—is detecting and managing the harmful algal blooms that appear on the river.
An experimental floating wetland and new research and analysis of water quality data that shows a possible effective detection system for algal blooms on the Charles River are two new steps toward the goal of safe, accessible swimming.Read More
The Gulf Stream, the massive western boundary current off the east coast of North America, moves water from the Gulf of Mexico north and west across the Atlantic Ocean. There’s a lot of energy in that much moving water and researchers are trying to put it to use.
Although the Gulf Stream’s path shifts (researchers say it acts like a wiggling garden hose), in a couple of spots, it stays relatively stable. At one such spot off the coast of Cape Hatteras, North Carolina, researchers have dropped moorings and research instruments to study the current with the eventual goal of harnessing it for renewable energy.Read More