YSI EXO Sonde Tune-Up Service
- Tune-up rate for the EXO1 and EXO2
- Sondes are returned with a Calibration Report to confirm that all parameters meet specifications
- Fondriest Environmental is a factory-authorized YSI Service and Repair Center
|Tune-Up EXO||Tune-up YSI EXO water quality sonde, includes all repair labor|
|Usually ships in 1-2 weeks|
|599594||EXO sonde tool kit|
|599680||EXO1 replacement O-ring kit|
|599473||EXO1 replacement bail|
|Usually ships in 3-5 days|
|599676||EXO central wiper brush guard kit, pack of 2|
|577603-01||EXO guarded pH sensor replacement module|
|577603-02||EXO unguarded pH sensor replacement module|
|577613-01||EXO guarded pH/ORP sensor replacement module|
|577613-02||EXO unguarded pH/ORP sensor replacement module|
|599110||EXO optical dissolved oxygen sensor replacement cap|
|599475||EXO bulkhead connector port plug|
|Usually ships in 3-5 days|
|599665||EXO female dummy plug kit|
Quality data can be directly correlated to the condition of your monitoring equipment. Instrument check-up, characterization, and certification by a factory-authorized service center are recommended on a regular basis to help ensure data accuracy and maximize the instrument’s usable life.
Fondriest Environmental offers excellent turnaround times and low service costs on YSI 6-Series sondes. Tune-Up Service covers all repair labor in addition to:
- Seal Inspection & Cleaning
- Cable Inspection & Testing
- Sensor Reconditioning & Calibration
- Circuit Board Performance & Upgrade
Upon receipt of the instrument, repair technicians will perform a free evaluation and send a detailed evaluation report and price quote via email to the point of contact.
Repairs are based on customer feedback and are performed only after approval is received. All repaired instruments are returned with a Calibration Report to confirm that system parameters meet factory specifications.
Call our Service & Repair Department at 888.426.2151 or email firstname.lastname@example.org for more information.
- Seals are cleaned of any contamination that may cause leakage
- Damaged or missing o-rings are replaced
- The cable is subject to a vigorous flex test to check for shorts
- Connector and ports are cleaned of any moisture and corrosion
- Sensors are reconditioned and calibrated so they are ready for use
- New DO membranes are installed with fresh electrolyte solution
- Circuit board performance is verified
- Software is upgraded to the latest version
- Calibration Report confirms that all parameters meet specifications
In The News
The Charles River used to be a swimming hotspot for Cambridge and Boston residents.
Decades of industrial pollution and nutrient runoff have degraded water quality and eliminated public swimming in the Lower Charles, but a movement is afoot to get Boston and Cambridge back in the water. One step toward the goal of a safely swimmable river—without the need to obtain a permit, as is now necessary—is detecting and managing the harmful algal blooms that appear on the river.
An experimental floating wetland and new research and analysis of water quality data that shows a possible effective detection system for algal blooms on the Charles River are two new steps toward the goal of safe, accessible swimming.Read More
The Gulf Stream, the massive western boundary current off the east coast of North America, moves water from the Gulf of Mexico north and west across the Atlantic Ocean. There’s a lot of energy in that much moving water and researchers are trying to put it to use.
Although the Gulf Stream’s path shifts (researchers say it acts like a wiggling garden hose), in a couple of spots, it stays relatively stable. At one such spot off the coast of Cape Hatteras, North Carolina, researchers have dropped moorings and research instruments to study the current with the eventual goal of harnessing it for renewable energy.Read More
In early 2020, Michigan found itself facing one of the worst outbreaks of Covid-19 in the country. Though it’s close to second nature now, businesses, schools and governments were suddenly forced to conduct business without close contact. Universities and research institutions had to pause some scientific research. Whatever was able to continue slowed to a crawl.
Around the Great Lakes, a network of buoys monitors dozens of water quality parameters and lake conditions, reporting them in real time. This year, the monitoring season was cut a bit short as Covid-19 restrictions hit in the weeks before buoys were set to be deployed.Read More