Hach sensION+ 9663 Ammonium Ion Selective Electrode
- Utilizes a unique solid-state sensor technology, eliminating costly membrane replacements
- Requires little maintenance
- Provides fast, stable, and accurate response in a variety of sample types
|LZW9663.97.0002||sensION+ 9663 ammonium ion selective electrode (ISE), sensor only|
|Usually ships in 3-5 days|
The Hach sensION+ 9663 ammonium ion selective electrode is a half-cell (reference not integrated) ammonium ion selective electrode (ISE) with an epoxy body and non-replaceable, solid-state PVC (polymer) membrane selective to ammonium ions in solution. It is recommended for use with the sensION+ 5044 reference electrode. The 9663 has a fixed 1 meter cable and BNC connector for laboratory use with the Hach sensION+ MM374 Multimeter. The 9663 is ideal for measuring ammonium concentrations in drinking water, wastewater and general water quality applications.
The Hach sensION+ 9663 ammonium ion selective electrode utilizes a unique solid-state sensor technology that eliminates costly membrane replacements, whereas traditional PVC-membrane ISEs require membrane replacement every 2-3 months. The 9663 ISE’s unique solid-state sensor technology eliminates the need for frequent membrane replacement by using a solid gel ion exchange behind the ion-selective membrane versus the liquid ion exchange used in most ISEs. This design allows users to get up to 2 years of life from their ISE without membrane replacement – saving the time and costly expense of frequently exchanging membranes.
The solid-state sensor design requires little maintenance and allows for dry storage of the ISE without a shelf life or membrane replacement. Ultimately, the 9663 provides fast, stable and accurate response in a variety of sample types.
- Dimensions (D x L): 12 mm x 120 mm
- Material Sensor Body: ABS
- Parameter: Ammonium
- Temperature Range: 5 to 40 °C
In The News
In Saskatoon, Saskatchewan, pollution and runoff from storms and snowmelt are getting the close look they deserve, and there’s much more to examine.
Weather, from heavy spring storms to long months of snow and freezing temperatures, makes the polluting potential of runoff and snowmelt greater than and different from warmer climate cities, said Garry Codling in an email. In Saskatoon, potentially harmful elements in runoff can exceed the guidelines for runoff set by the Canadian government.Read More
Appalachia may be as closely associated with mining as it is to anything else. That close relationship will leave its mark on the area’s streams long after the last mine closes.
A nine-year study recently published in Science of the Total Environment shows that long after mining activity stops and the land is left to heal, streams and stream life are slow to recover.
“We could be really fine point and say that some of them seem to be recovering very, very slowly,” said Carl Zipper, professor emeritus of environmental science at Virginia Tech University . Most of the streams studied didn’t show signs of recovery.Read More
An unusual nuisance is slowly growing into an inexplicable problem for researchers at Montana’s Department of Environmental Quality .
For the last five years, a native species of algae called Cladophora has covered large portions of the Smith River, one of the state’s most popular waterways for boating, fishing and recreating. And scientists don’t know why.
“It’s just unusual. I don’t know if it’s extreme for the state of Montana as other systems have had Cladophora problems as well. But it’s most unusual due to the lack of land use changes,” said Chace Bell, a water quality assessment specialist with the Montana DEQ.Read More