Hach Sodium ISE Analysis Package
- Special doping agents in the glass improve selectivity for sodium ions
- Improved linearity, response speed, and conditioning time
- Renewable, free flowing reference junction for fast and accurate sodium measurements
|2348100||Sodium ISE analysis package includes Hach sodium platinum series combination electrode, (2) ammonium chloride reference cartridges, 100 and 1000 mg/L standards, reagents, beaker, and manual|
|Usually ships in 3-5 days|
The electrode uses an improved sodium-selective glass membrane containing an inner filling solution with a constant sodium ion concentration and glycerol as an antifreeze agent. The
glass is a LAS (lithium aluminosilicate) glass. Special doping agents in the glass improve selectivity for sodium ions. A potential develops across the membrane that is proportional to the
sodium concentration in the standard or sample.
- Range: 0.010 mg/L Na+ to saturated solution
- pH Range: from 4 to 14
- Slope: 58 +/-3 mV at 25 C
- Linear Region: 1 mg/L to saturation
- Temp Range: 10 to 30 C (routine use), 10 to 50 C (occasional use), -40 to 50 C (storage)
- Response Time: Below 1.0 mg/L, 5-10 minutes; above 1.0 mg/L, 1-2 minutes
- Minimum Sample Volume: 25 mL
- Electrode Resistance: Less than 1400 Mohms at 30 C (new)
- Tip Diameter: 12 mm
- Tip Length: 115 mm
- Total Length: 240 mm with dispenser
- Cable Length: 1.0 m (36 in.)
- Cable Connector: BNC
- (1) Sodium platinum series combination electrode
- (2) Ammonium chloride reference cartridges
- (1) 1000mL bottle of sodium standard, 100 mg/L
- (1) 500mL bottle of sodium standard, 1000 mg/L
- (100) Sodium ISA powder pillows
- (1) Sodium chloride, ACS, 454 g
- (1) 50mL beaker
- (1) Operations manual
In The News
Is eradicating Great Lakes sea lamprey an “impossible dream?” Researchers say no
The sea lamprey’s days in the Great Lakes could be numbered.
That’s according to one researcher who took one of the first scientific looks at the possibility of sea lamprey eradication in the Great Lakes.
So, can you remove enough sea lamprey to make them disappear?
“Well the answer is we already have,” said Michael Jones, emeritus professor of fisheries and wildlife at Michigan State University. “Then there’s the obvious question: Why are they still here?”
While multiple gaps in current management techniques, like sea lamprey poisons called lampricides, could account for sea lamprey’s persistence in the Great Lakes, new technology could help sea lamprey managers eliminate inaccessible populations.Read More
The Shasta crayfish and signal crayfish are two similar looking arthropods on two very different ecological trajectories. As one spreads in abundance, originating in the Pacific Northwest and spreading throughout the world, the other has been reduced to a handful of remaining populations spread throughout one river and its tributaries.
Pacifastacus leniusculus - the signal crayfish - has met few obstacles in its widely successful expansion from the Pacific Northwest southward in California and Nevada, as well as Europe and Japan. By some expert accounts, it has reached invader status. And while invasive species are rarely good for the surrounding food webs, it’s Pacifastacus fortis - the Shasta crayfish - that’s suffered the most at the signal crayfish’s fortune.Read More
What might the Maine Aquaculture Innovation Center’ s (MAIC) buoy offer that other governments and university monitoring equipment lack? The center doesn’t have MicroCAT recorders or autonomous acoustic sensing gliders. It’s not deploying hundred-thousand-dollar oceanographic mooring lines gathering massive amounts of data.
So what can MAIC’s three-foot prototype buoy offer that others can’t? It’s easy to clean and costs very little.
“One of the big issues for putting anything in the water is biofouling,” said Josh Girgis, an engineer at MAIC based at the University of Maine’s Darling Marine Center (DMC). “If you put a sensor in, you can only expect it to work until something starts growing on it.Read More