Used YSI 600OMS V2 Optical Monitoring Sonde
- Compact sonde, easily fits in 2-inch wells
- Wiped optics and optional battery pack for long term deployments
- Compatible with NexSens real-time data logging systems
|600-01-R||Used 600OMS V2 Sonde with temperature/conductivity sensor|| |
|600-02-R||Used 600OMS V2 Sonde with temperature/conductivity sensor, internal battery pack|| |
|6136-R||Used 6136 turbidity sensor with self-cleaning wiper|| |
|6130-R||Used 6130 Rhodamine WT sensor with self-cleaning wiper|
Designed for use in fresh, sea or polluted waters, the YSI 600OMS V2 utilizes the field-proven YSI sensors and incorporates innovations in sensor configuration such as a conductivity and temperature module that fits into the sonde body.
Optical sensor options include optical dissolved oxygen, turbidity, chlorophyll, blue-green algae (both phycocyanin and phycoerythrin), and rhodamine. All optical sensors have built-in wipers that activate prior to sensor readings. Combined with depth or vented level, the 600OMS V2 is a powerful sampling tool.
The 600OMS V2 is available with or without internal power. Its small size is perfect for applications such as turbidity or oxygen monitoring.
- Medium: Fresh, sea, or polluted water
- Operating Temperature: -5 to +50 C
- Storage Temperature: -10 to +60 C
- Communications: RS-232, SDI-12
- Software: EcoWatch
- Diameter: 1.65" (4.19cm)
- Length: 21.3" (54.1cm)
- Internal Power: 4 AA-size alkaline batteries
- External Power: 12 VDC
- (1) YSI 600OMS Sonde
- (1) Temperature/conductivity sensor
- (1) Soft-sided carrying case
- (1) EcoWatch for Windows software CD
- (1) Calibration cup
- (1) Probe guard
- (1) 6-Series operations manual
- (1) Maintenance kit
The used YSI 600OMS Sonde has a 90 day Warranty.
Used products do not come from trade-ins, they come from Fondriest Rental Pool.
In The News
Water quality issues are shifting in the United States’ rivers in big ways.
Those changes are driven, in part, by the way the land in a watershed is used and they’re big enough that researchers may need to change the way they think about water quality in the American rivers.
“What was striking to us was how perceptions of water quality issues from several decades ago may need to be updated,” said Edward Stets, a U S Geological Survey research ecologist, in an email response to questions from Environmental Monitor.
New research by Stets published in Environmental Science &; Technology in March highlights these shifting water quality issues.Read More
The start of Duke University’s oceanographic mooring line doesn’t begin at the surface of the ocean, but 500 meters beneath it. Floating at the top of the mooring system is a 64-inch syntactic sphere with 2,500 Lbs of buoyancy. It serves a duo of roles as both a floatation device and housing for two high-end acoustic monitoring systems.
Next comes 75 meters of chains and wires before coming to two ocean current profilers, one pointing up and the other pointing down. Next to that equipment are two instruments that measure temperature and salinity.
Next comes several hundred more meters of wire before arriving at several glass balls, more floatation orbs.Read More
This spring, palm warblers, rose-breasted grosbeaks, tree swallows and great blue herons fill Muskegon Lake Nature Preserve with their songs or stately presence.
In 1997, the preserve was a dump.
The community partnerships and restoration efforts that turned a dump in one corner of Muskegon Lake in west Michigan into a haven for plants, wildlife and recreational fishers ran parallel to other efforts around the lakeshore and the Great Lakes since 1985.
In 1985, the United States and Canada chose to focus environmental cleanup efforts on 42 highly polluted areas of concern (AOCs ) around the Great Lakes, including Muskegon Lake.
Thirty-five years later, that decision has changed environmental remediation and benefited Great Lakes communities.Read More