YSI 6025 Chlorophyll Sensor
- Fast, convenient, in-situ monitoring
- Self-wiping probe for long-term measurement
- Small field-replaceable probe design
|606025||6025 chlorophyll sensor with self-cleaning wiper|| |
|600-01||600OMS V2 Sonde with temperature/conductivity sensor|| |
|Usually ships in 3-5 days|
|606144||6144 optical probe wiper pad kit, 20 pack of wiper pad strips|
|106023-01P||FWT 25 Rhodamine WT dye, 2.5% active ingredient, 1 pint|
|606624||6624 optical wiper kit, 2 pack, for use with YSI 6025 & 6130 optical probes|
The YSI 6025 Chlorophyll Probe uses technology similar to that used by fluorometers, but the YSI sensor is much smaller, making it compatible with the probe ports in YSI's 6-Series instruments, including the YSI 6600 & 6600 V2 as well as the YSI 6820, 6820 V2, 6920, & 6920 V2. The YSI 6600 V2 sonde allows simultaneous measurement of four optical probes (dissolved oxygen, turbidity, chlorophyll, blue-green algae, and/or rhodamine), along with other parameters. The YSI 6820 V2 and 6920 V2 allow measurement of two optical probes, along with many other parameters.
Chlorophyll in various forms is bound within the living cells of algae, phytoplankton, and other plant matter found in water. Chlorophyll is a key component in the process of photosynthesis, the critical process in which energy from sunlight is used to produce life-sustaining oxygen. The amount of chlorophyll in a water sample is used as a measure of suspended phytoplankton, the magnitude of which can significantly affect the overall quality of water.
Before the new YSI sensor made in situ measurement convenient, the common way to measure chlorophyll was to collect samples and use extractive analysis in a laboratory, or take large equipment to the field. Extractive analysis methods, though highly accurate, are time-consuming and require an experienced analyst. The YSI optical, fluorescence probe is quick and efficient to use, and enables spot sampling in remote areas as well as continuous or interval monitoring.
- Range: ~0 to 400 ug/L; 0 to 100 RFU
- Detection Limit: ~0.1 ug/L
- Resolution: 0.1 ug/L Chl; 0.1% RFU
- Linearity: R2> 0.9999
- Warranty: 2 years
During unattended studies, the stability can be improved by increasing sampling frequency. To change the frequency settings, Go to Advanced Menus-->data filter-->frequency-->time constants and change the time constant to 30 seconds.
Yes, the wiper should be run at least once during calibration. The calibration reading should not be accepted until at least 30 seconds after a wipe.
In The News
Zooplankton drifts through the ocean, often ignored by the public in favor of more charismatic marine organisms farther up the food chain.
A new crowdsourced project aims to change that, giving anyone a closer look at small and intricate zooplankton. PlanktonPortal.org features 900,000 high resolution images of zooplankton and tasks the public to identify them as part of data processing for a collaborative study, based out of the University of Miami Rosenstiel School of Marine and Atmospheric Science.
Researchers hope the project will tell them more about zooplankton behavior, grouping and interaction.Read More
Ohio State University’s Stone Laboratory will open a new lab this summer focused on algae and water quality, according to the Ohio Sea Grant magazine Twine Line .
The new laboratory will be in the Stone Lab Research Building on Lake Erie’s South Bass Island.
Many instruments and tests will be accessible at the lab to evaluate all functions of algal blooms from photosynthesis to nutrient consumption. The lab will enable scientists to test phytoplankton, water and sediment samples on the lake instead of sending them away. This should save researchers time and money in the long run.
Lab administrators hope to increase collaborations with other research organizations and government agencies through the creation of the new laboratory.Read More
Contaminated stormwater threatens a lot of water in the United States.
Nearly 50,000 miles of rivers, 760,000 acres of wetlands and one million acres of estuaries are threatened by contaminated stormwater, according to the US Environmental Protection Agency.
Those numbers are cited in a review of research recently published in Environmental Science: Water Resource &; Technology that looks at one tool for tackling that threat: biochar-augmented biofilters.Read More