The Self-contained Zebra-Tech LI-COR LI-192 Hydro-Wiper is a field-proven, high performance wiping system designed for the LI-COR LI-192 submersible PAR sensors.
|Image||Part #||Product Description||Price||Stock||Order|
|H-SC-001-030||Self-contained Hydro-Wiper for LI-COR LI-192 PAR sensor, 30m depth rating||
|Usually ships in 1-2 weeks|
How can I mount my wiper control housing?
The marine grade (316) stainless steel bracket attached to the wiper control housing can attach the housing to any required structure. Avoid attaching directly to a metal structure, as electrolysis can occur which may result in rapid deterioration of the metal parts.
How will I know when my Hydro-Wiper is low on battery?
When the battery reaches around 6.5 volts, the wipes will cease. The LED indicator will blink three times every 15 seconds when the Hydro-Wiper is in low battery shutdown mode.
How often should I set my Hydro-Wiper to wipe?
The optimal wipe interval will depend on the environmental conditions at the field site. Generally a wipe interval of 2 to 3 hours should prove sufficient. The wipe interval is controlled by the position of the wipe interval switch. The wipe interval switch can be set using the screwdriver provided in the field kit. Each position adds 15 minutes to the wipe interval, ranging from 15 to 720 minutes.
For decades, scientists have been fascinated by the color and clarity of water. Very high water clarity is extremely rare in nature. So it is was with some surprise that a team of scientists from the National Institute of Water and the Atmosphere in New Zealand recently discovered that Blue Lake, in Nelson Lakes National Park on the country’s South Island, is the visually clearest freshwater lake ever reported. Mark Gall, a scientist at NIWA, said colleague Rob Merrilees had seen Blue Lake and thought it could be comparable to Pupu Springs, another South Island lake that was the clearest reported freshwater at the time.Read More
The world is an unpredictable place full of fouling and entanglements that can send just about anyone astray. Water quality data too can be severely compromised by unforgiving conditions and unpredictable events. John Radford, Zebra-Tech’s founder, learned firsthand how fouling could degrade data quality to the point of being unusable. “Whilst working for the National Institute of Atmospheric Research in New Zealand, we had constant issues trying to make long-term measurements of suspended sediment concentrations in shallow coastal water,” he said. “Fouling of the sensors between site visits caused fatal degradation of the data quality. Attempts were made to develop a method for mathematically correcting the data, but these were unsuccessful.Read More
Thinking of hitting the ice with a SondeCAM underwater fishing camera? Due to its rugged design, you won't have to worry about it handling the harsh elements. However there are a few simple tricks to get the most out of a FishSens SondeCAM while ice fishing. You won't have to do anything to modify the SondeCAM itself, but you are going to have to bring a few extra things. Most importantly we are going to need a power source. Unless you are hauling your gear with a truck, you'll want something more portable than the battery you used in the boat. Pick up an inexpensive and maintenance-free 12-volt, 9-amp battery. It is going to provide plenty of power, but will be much lighter and take up less space.Read More