The OTT RLS, non-contact radar level sensor with pulse radar technology is ideal for monitoring in remote or hard to reach locations.
The RLS non-contact radar level sensor with pulse radar technology is ideal for monitoring in remote areas and applications where conventional measuring systems are not suitable. The RLS accurately and efficiently measures surface water level With a non-contact distance range of up to 115 feet above water. The sensor is IP67 waterproof and has extremely low power consumption, making it ideal for solar-charged monitoring systems.
The radar level sensor uses a revolutionary level measurement technology, meeting the USGS accuracy requirement of +/-0.01 feet. Two antennas are enclosed in a compact housing and transmit pulses toward the water surface. The time delay from transmission to receipt is proportional to the distance between sensor and water surface. A sampling rate of 16 Hz (16 measurements/second) with 20 second averaging minimizes water surface conditions such as waves and turbulence. The RLS does not require calibration and is unaffected by air temperature, humidity, flood events, floating debris, or contaminated water.
|Image||Part #||Product Description||Price||Stock||Order|
|63.107.001.9.2S||RLS radar water level sensor, FCC Version (25 GHz), SDI-12 & 4-20mA output||Usually ships in 1-2 weeks|
Where should I place my Radar Water Level Sensor?
In order to obtain reliable and correct measurements, you should avoid submerged obstructions and there should be a clear path between the sensor and the water to avoid false reflections. There is a full list of helpful site selection tips under the Documents Tab.
What maintenance do I need to preform to keep my Radar Water Level Sensor clean?
The OTT RLS sensor does not require calibration and there are no parts that need regular replacing. To maintain your sensor be sure to: check for dirt, obstructions in the measurement beam, and plausibility of the measured values. To clean your sensor, use commercial, gentle and non-erasing cleaners with a soft sponge.
My sensor is not responding to the SDI-12 Interface, what should I check?
There are a few possibilities when having trouble with the SDI-12 interface. First, check the fuse in the power supply. If that is working, the power supply may be too low or too high. You should be using 12/24V. If your voltage is correct, make sure you are using DC (direct current).
Growing from a 38-acre purchase in 1998 to 298 acres in 2004 to the 305 acres it encompasses today; the Black Fork River Wetlands features habitats not found just anywhere, including buttonbush swamp, swamp forest, marsh, riparian corridor and uplands habitats. Beavers make their homes there, as well as trumpeter swans, bald eagles, soras and sandhill cranes.
While it may seem picturesque and undisturbed, it is in fact embattled due to human activity on all sides. “It’s a multi-use area,” says Jenna Binder, a visiting Assistant Professor in Ashland University’s Biology and Toxicology Department. “It’s strongly influenced by the heavy agriculture in this area of Ohio. Oil and gas industry fracking is also being done in the area.Read More
Biological field stations make it possible for researchers all over the country to conduct environmental research. While some field stations have artist residencies, art is typically not the main focus of the biological station. Not so at Bakersville, North Carolina’s new AS IF Center (Art + Science In The Field) , which just opened its doors in March 2018. At AS IF, researchers and artists are deliberately invited to commingle, collaborate and create new things together. Far from being on the periphery or existing as an afterthought, artists are considered to be on parity with researchers at AS IF, the one energized by the other’s perspective.Read More
The polar regions of the world have always a challenge for scientists to explore and study. Even logistics that are typically no more than passing concerns under other circumstances such as transportation become major problems during polar wintertime. Now, r esearchers are reporting on their use of hundreds of oceanic floats that are drifting and diving their way through the Southern Ocean, including under its ice, with surprising results.
Happy robotic wanderers
EM spoke with Dr. Alison Gray , assistant professor of physical oceanography at the University of Washington , to find out more about the work, the robots, and the significance of the findings in improving our understanding of the global climate and this poorly studied region.Read More