OTT RLS Radar Water Level Sensor
- Transmit & receive antenna enclosed in a lightweight, durable housing with flat antenna design
- Easily mounts to a bridge, frame, pipeline, or extension arm
- Connects to NexSens X2 data logging system via SDI-12 interface
|6310900192S||RLS radar water level sensor, FCC Version (25 GHz), SDI-12 & 4-20mA output|| |
|Usually ships in 1-2 weeks|
|C8P-24-P||8 conductor 24 AWG cable, PVC jacket, priced per ft.|
|Usually ships in 3-5 days|
|UW-FL3||UW plug to flying lead cable, 3m|
|Usually ships in 3-5 days|
The RLS non-contact radar level sensor with pulse radar technology is ideal for monitoring in remote areas and applications where conventional measuring systems are not suitable. The RLS accurately and efficiently measures surface water level With a non-contact distance range of up to 115 feet above water. The sensor is IP67 waterproof and has extremely low power consumption, making it ideal for solar-charged monitoring systems.
The radar level sensor uses a revolutionary level measurement technology, meeting the USGS accuracy requirement of +/-0.01 feet. Two antennas are enclosed in a compact housing and transmit pulses toward the water surface. The time delay from transmission to receipt is proportional to the distance between sensor and water surface. A sampling rate of 16 Hz (16 measurements/second) with 20 second averaging minimizes water surface conditions such as waves and turbulence. The RLS does not require calibration and is unaffected by air temperature, humidity, flood events, floating debris, or contaminated water.
- (1) Radar level sensor
- (1) 2-part swivel mount
- (1) Installation kit - Includes (4) 6x40mm wood screws & (4) plastic plugs
- (2) Double open-ended wrenches (10x13)
- (1) Factory acceptance test certificate (FAT)
- (1) Operations manual
Yes, the sensor does not have logging capabilities and needs to be integrated with a data logger. Sensor output options are SDI-12, SDI-12 via RS-485 and 4-20mA.
While ultrasonic sensors emit high frequency (20 kHz to 200 kHz) acoustic waves, radar sensors use radio-frequency signals (1GHz to 60 GHz) and readings are generally less affected by pressure, temperature and moisture changes.
The radar level sensor is mounted to a structure so that the radar beam is perpendicular to the surface of the water. A guide for installation and site selection can be found here: https://www.fondriest.com/pdf/ott_rls_install.pdf
In The News
The Charles River used to be a swimming hotspot for Cambridge and Boston residents.
Decades of industrial pollution and nutrient runoff have degraded water quality and eliminated public swimming in the Lower Charles, but a movement is afoot to get Boston and Cambridge back in the water. One step toward the goal of a safely swimmable river—without the need to obtain a permit, as is now necessary—is detecting and managing the harmful algal blooms that appear on the river.
An experimental floating wetland and new research and analysis of water quality data that shows a possible effective detection system for algal blooms on the Charles River are two new steps toward the goal of safe, accessible swimming.Read More
The Gulf Stream, the massive western boundary current off the east coast of North America, moves water from the Gulf of Mexico north and west across the Atlantic Ocean. There’s a lot of energy in that much moving water and researchers are trying to put it to use.
Although the Gulf Stream’s path shifts (researchers say it acts like a wiggling garden hose), in a couple of spots, it stays relatively stable. At one such spot off the coast of Cape Hatteras, North Carolina, researchers have dropped moorings and research instruments to study the current with the eventual goal of harnessing it for renewable energy.Read More
In early 2020, Michigan found itself facing one of the worst outbreaks of Covid-19 in the country. Though it’s close to second nature now, businesses, schools and governments were suddenly forced to conduct business without close contact. Universities and research institutions had to pause some scientific research. Whatever was able to continue slowed to a crawl.
Around the Great Lakes, a network of buoys monitors dozens of water quality parameters and lake conditions, reporting them in real time. This year, the monitoring season was cut a bit short as Covid-19 restrictions hit in the weeks before buoys were set to be deployed.Read More