YOUNG ResponseONE Weather Transmitter
- Measures four key meteorological variables with integrated compass
- Serial output formats include SDI-12, NMEA, and ASCII text
- Wiring connections are made in a convenient weather-proof junction box
|92000||ResponseONE weather transmitter|
|Usually ships in 1-2 weeks|
|18660||Sensor cable, 8 conductor shielded, 22 AWG, per ft.|
|Drop ships from manufacturer|
|Drop ships from manufacturer|
The YOUNG ResponseONE Weather Transmitter measures four key meteorological variables with one compact instrument. It is ideal for many weather monitoring applications requiring accurate, and reliable measurements. Ultrasonic wind speed and direction, atmospheric pressure, humidity and temperature sensors are carefully integrated into an enclosure optimized for durability, airflow and mitigation of solar radiation effects. An integrated compass helps enable mobile applications. A variety of useful serial output formats are provided including SDI-12, NMEA, and ASCII text. Output may be continuously provided or, to conserve power, polled output may be used. RS-232 or RS-485 serial format options enable direct integration with YOUNG displays, marine NMEA systems, data loggers or other compatible serial devices. An easy-to-use Windows setup program is provided with each sensor. The program allows the user to customize the device settings such as sampling rates and communication parameters.
The ResponseONE features durable, corrosion-resistant construction and installs on readily available 1 inch (IPS) pipe. Wiring connections are made in a convenient weather-proof junction box. Special connectors and cables are not required.
Range: 0–70 m/s (156mph)
Resolution: 0.01 m/s
±2% or 0.3 m/s (0–30m/s)
±3% (30 – 70 m/s)
Azimuth Range: 0-360 degrees
Resolution: 0.1 degree
Accuracy: ±2 degrees
Range: -40 to +60°C
Range: 500–1100 hPa
Resolution: 0.1 hPa
Accuracy: ±0.5 hPa
Range: 0–360 degrees
Resolution: 1 degree
Accuracy: ±1.4 degrees
Serial Output (selectable):
Interface: RS-232, RS-485/422, SDI-12
Formats: NMEA, SDI-12, ASCII (polled or continuous)
Baud Rates: 1200, 4800, 9600, 19200 and 38400
Voltage: 10–30 VDC
Current: 7 mA @ 12 VDC typical, 80 mA max
Protection Class: IP65
EMC Compliance: FCC Class A digital device, IEC Standard 61326-1
Dimensions: 30 cm high x 13.5 cm wide
Weight: 0.7 kg (1.5lb)
Shipping Weight: 1.6 kg (3.5lb)
Operating Temperature: -40 to +60°C
Removable Bird Spikes: Included
In The News
In Saskatoon, Saskatchewan, pollution and runoff from storms and snowmelt are getting the close look they deserve, and there’s much more to examine.
Weather, from heavy spring storms to long months of snow and freezing temperatures, makes the polluting potential of runoff and snowmelt greater than and different from warmer climate cities, said Garry Codling in an email. In Saskatoon, potentially harmful elements in runoff can exceed the guidelines for runoff set by the Canadian government.Read More
Appalachia may be as closely associated with mining as it is to anything else. That close relationship will leave its mark on the area’s streams long after the last mine closes.
A nine-year study recently published in Science of the Total Environment shows that long after mining activity stops and the land is left to heal, streams and stream life are slow to recover.
“We could be really fine point and say that some of them seem to be recovering very, very slowly,” said Carl Zipper, professor emeritus of environmental science at Virginia Tech University . Most of the streams studied didn’t show signs of recovery.Read More
An unusual nuisance is slowly growing into an inexplicable problem for researchers at Montana’s Department of Environmental Quality .
For the last five years, a native species of algae called Cladophora has covered large portions of the Smith River, one of the state’s most popular waterways for boating, fishing and recreating. And scientists don’t know why.
“It’s just unusual. I don’t know if it’s extreme for the state of Montana as other systems have had Cladophora problems as well. But it’s most unusual due to the lack of land use changes,” said Chace Bell, a water quality assessment specialist with the Montana DEQ.Read More