YOUNG ResponseONE Weather Transmitter
- Measures four key meteorological variables with integrated compass
- Serial output formats include SDI-12, NMEA, and ASCII text
- Wiring connections are made in a convenient weather-proof junction box
|92000||ResponseONE weather transmitter|
|Usually ships in 1-2 weeks|
|18660||Sensor cable, 8 conductor shielded, 22 AWG, per ft.|
|Drop ships from manufacturer|
|Drop ships from manufacturer|
The YOUNG ResponseONE Weather Transmitter measures four key meteorological variables with one compact instrument. It is ideal for many weather monitoring applications requiring accurate, and reliable measurements. Ultrasonic wind speed and direction, atmospheric pressure, humidity and temperature sensors are carefully integrated into an enclosure optimized for durability, airflow and mitigation of solar radiation effects. An integrated compass helps enable mobile applications. A variety of useful serial output formats are provided including SDI-12, NMEA, and ASCII text. Output may be continuously provided or, to conserve power, polled output may be used. RS-232 or RS-485 serial format options enable direct integration with YOUNG displays, marine NMEA systems, data loggers or other compatible serial devices. An easy-to-use Windows setup program is provided with each sensor. The program allows the user to customize the device settings such as sampling rates and communication parameters.
The ResponseONE features durable, corrosion-resistant construction and installs on readily available 1 inch (IPS) pipe. Wiring connections are made in a convenient weather-proof junction box. Special connectors and cables are not required.
Range: 0–70 m/s (156mph)
Resolution: 0.01 m/s
±2% or 0.3 m/s (0–30m/s)
±3% (30 – 70 m/s)
Azimuth Range: 0-360 degrees
Resolution: 0.1 degree
Accuracy: ±2 degrees
Range: -40 to +60°C
Range: 500–1100 hPa
Resolution: 0.1 hPa
Accuracy: ±0.5 hPa
Range: 0–360 degrees
Resolution: 1 degree
Accuracy: ±1.4 degrees
Serial Output (selectable):
Interface: RS-232, RS-485/422, SDI-12
Formats: NMEA, SDI-12, ASCII (polled or continuous)
Baud Rates: 1200, 4800, 9600, 19200 and 38400
Voltage: 10–30 VDC
Current: 7 mA @ 12 VDC typical, 80 mA max
Protection Class: IP65
EMC Compliance: FCC Class A digital device, IEC Standard 61326-1
Dimensions: 30 cm high x 13.5 cm wide
Weight: 0.7 kg (1.5lb)
Shipping Weight: 1.6 kg (3.5lb)
Operating Temperature: -40 to +60°C
Removable Bird Spikes: Included
In The News
The Charles River used to be a swimming hotspot for Cambridge and Boston residents.
Decades of industrial pollution and nutrient runoff have degraded water quality and eliminated public swimming in the Lower Charles, but a movement is afoot to get Boston and Cambridge back in the water. One step toward the goal of a safely swimmable river—without the need to obtain a permit, as is now necessary—is detecting and managing the harmful algal blooms that appear on the river.
An experimental floating wetland and new research and analysis of water quality data that shows a possible effective detection system for algal blooms on the Charles River are two new steps toward the goal of safe, accessible swimming.Read More
The Gulf Stream, the massive western boundary current off the east coast of North America, moves water from the Gulf of Mexico north and west across the Atlantic Ocean. There’s a lot of energy in that much moving water and researchers are trying to put it to use.
Although the Gulf Stream’s path shifts (researchers say it acts like a wiggling garden hose), in a couple of spots, it stays relatively stable. At one such spot off the coast of Cape Hatteras, North Carolina, researchers have dropped moorings and research instruments to study the current with the eventual goal of harnessing it for renewable energy.Read More
In early 2020, Michigan found itself facing one of the worst outbreaks of Covid-19 in the country. Though it’s close to second nature now, businesses, schools and governments were suddenly forced to conduct business without close contact. Universities and research institutions had to pause some scientific research. Whatever was able to continue slowed to a crawl.
Around the Great Lakes, a network of buoys monitors dozens of water quality parameters and lake conditions, reporting them in real time. This year, the monitoring season was cut a bit short as Covid-19 restrictions hit in the weeks before buoys were set to be deployed.Read More