YSI EXO Conductivity & Temperature Sensor
- 0 to 200 mS/cm measurement range
- T63<2 sec response time
- ±0.5% of reading or 0.001 mS/cm accuracy from 0 to 100
|599870||EXO conductivity & temperature sensor|| |
|065270||3161 conductivity standard, 1,000 uS, 1 quart|
|065272||3163 conductivity standard, 10,000 uS, 1 quart|
|065274||3165 conductivity standard, 100,000 uS, 1 quart|
|Usually ships in 3-5 days|
|060907||3167 conductivity standard, 1,000 uS, 8 pints|
|060911||3168 conductivity standard, 10,000 uS, 8 pints|
|060660||3169 conductivity standard, 50,000 uS, 8 pints|
|599867||EXO anti-fouling copper screen kit, pack of 2|
|599470||EXO conductivity/temperature sensor cleaning brush, pack of 2|
The EXO combination conductivity and temperature sensor should be installed in a sonde in nearly all sonde applications. Not only will this sensor provide the most accurate and fastest response temperature data, but it will also provide the best data for the use in temperature compensation for the other EXO probes. The conductivity data is used to calculate salinity, non-linear function (nLF) conductivity, specific conductance, and total dissolved solids, and compensate for changes in density of water (as a function of temperature and salinity) in depth calculations if a depth sensor is installed.
The temperature sensor uses a highly stable and aged thermistor with extremely low-drift characteristics. The thermistor’s resistance changes with temperature. The measured resistance is then converted to temperature using an algorithm. The temperature sensor receives a multi-point NIST traceable wet calibration and the accuracy specification of 0.01˚C is valid for expected life of the probe. No calibration or maintenance of the temperature sensor is required, but accuracy checks can be conducted.
The conductivity sensor uses four internal, pure-nickel electrodes to measure solution conductance. Two of the electrodes are current driven, and two are used to measure the voltage drop. The measured voltage drop is then converted into a conductance value in milliSiemens (millimhos). To convert this value to a conductivity value in milliSiemens per cm (mS/cm), the conductance is multiplied by the cell constant that has units of reciprocal cm (cm-1). The cell constant for the conductivity cell is approximately 5.5/cm ±10%. For most applications, the cell constant is automatically determined (or confirmed) with each deployment of the system when the calibration procedure is followed.
EXO sensors have internal thermistors for quality assurance purposes. Turbidity uses the internal thermistor for temperature compensation, while all other EXO sensors reference the C/T probe for temperature compensation. To display and log temperature, a C/T probe must be installed in an EXO sonde. Thermistor readings are logged in the sonde’s raw data–viewable in KOR software–but are not included in data exported to Excel.
In The News
The North Carolina Wildlife Resources Commission ’s Inland Fisheries Division has been working to restore brook trout in the state. Coldwater research coordinator Jacob Rash, who works with the brook trout team technicians on this project, spoke to EM about the work.
“In North Carolina, brook trout are our only native trout species,” explains Mr. Rash. “With that come biological and ecological considerations as well as cultural importance. A lot of folks here grew up fishing for brook trout with their relatives, so it's an important species that we work to try to conserve. We've done quite a bit of work to figure out where those brook trout populations are and what they are, in terms of genetics.Read More
Each year in Germany, as many as 450,000 living fish undergo live animal experiments to test how fish-friendly hydroelectric power plants in the country are. The idea is to discover how readily the fish can move through hydroelectric turbine installations in order to ultimately reduce mortality rates.
Of course, subjecting live fish to a potentially deadly test to save others is a bitter irony. And it's one that a team of scientists from the RETERO research project hopes to eventually mitigate with a robotic fish for testing.
EM corresponded with Olivier Cleynen and Stefan Hoerner from the University of Magdeburg about the complex flow conditions that set the parameters for the project.Read More
News stories about dogs getting sick from harmful algal blooms (HABs) in lakes have caused worry among members of the public this summer more than once. But Regional Science Consortium (RSC) Executive Director Dr. Jeanette Schnars and a dedicated team are bringing awareness about HABs to the public with the Mobile HAB Lab.
“We just launched the HAB Citizen Scientists program this year,” explains Dr. Schnars. “It helps us work with people, especially people who spend time at marinas frequently, that are out there all season long.”
The season for boaters at Presque Isle, where RSC is located, starts in mid-May and usually continues through the beginning or middle of October.Read More